Using gas chromatography: the model airTOXIC BTX PID measures the amount of Benzene Toluene Ethylbenzene and Xylenes in the air from sub-ppb levels up to 1.000 ppb.
Teledyne API 100 UV Fluorescent SO2 Analyser
Teledyne API 100A UV Fluorescent SO2 Analyser
Teledyne API 100E UV Fluorescent SO2 Analyser
Teledyne API 101 UV Fluorescent SO2 Analyser
Teledyne API 101A UV Fluorescent SO2/H2S Analyser
Teledyne API 200 chemiluminescent NOx analyser
Teledyne API 200A chemiluminescent NOx analyser
Teledyne API 200AU chemiluminescent NOx analyser
Teledyne API 200E chemiluminescent NOx analyser
Teledyne API 200EU chemiluminescent NOx analyser
Teledyne API 200EU chemiluminescent NOx analyser - blue light
Teledyne API 200UP chemiluminescent NOx analyser
Teledyne API 300 gas filter correlation CO analyser
Teledyne API 300A gas filter correlation CO analyser
Teledyne API 300E gas filter correlation CO analyser
Teledyne API 400 UV photometric O3 analyser
Teledyne API 400A UV photometric O3 analyser
Teledyne API 400E UV photometric O3 analyser
Teledyne API M100E UV Fluorescent SO2 Analyser
Teledyne API M200E chemiluminescent NOx analyser
Teledyne API M300E gas filter correlation CO analyser
Teledyne API M400E UV photometric O3 analyser
Teledyne API T100 UV Fluorescent SO2 Analyser
Teledyne API T100U UV Fluorescent SO2 Analyser
Teledyne API T200 chemiluminescent NOx analyser
Teledyne API T200U chemiluminescent NOx analyser
he Model T200U Trace-level NO/NO2/NOx analyzer has been developed specifically to address the challenges of low-level monitoring as required in the US NCore network. It uses the proven chemiluminescence principle and is designed to allow ultra-sensitive measurements with a lower detectable limit of 50 ppt while still meeting the requirements for use as a US EPA compliance analyzer.The Model T200U combines a gold plated reaction cell with an enhanced performance pump special low noise photomultiplier tube and unique signal conditioning to provide exceptional sensitivity. In addition a pre-reactor separates the NO-O3 reaction from background chemiluminescence to allow accurate auto-zero of the analyzer.All T Series instruments offer an advanced color display capacitive touch screen intuitive user interface flexible I/O and built-in data acquisition capability. All instrument set up control and access to stored data and diagnostic information is available through the front panel or via RS232 Ethernet or USB com ports either locally or by remote connection.The Model T200U comes with NumaView™ Software. NumaView™ Remote PC Software allows for a remote connection with virtual interface and data downloading capability to analyzers operating NumaView™ Software.e allows for a remote connection with virtual interface and data downloading capability to analyzers operating NumaView™ Software.
Teledyne API T300 gas filter correlation CO analyser
Teledyne API T400 UV photometric O3 analyser
The airmoVOC BTX is a gas chromatograph for the analysis and monitoring of trace amounts of benzene, toluene, ethylbenzene (m/p/o) xylene by FID detection.
The method applied by the APM-2 uses the light scattered by tiny particles (nephelometry) to determine the PM10 or PM2.5 concentration in the ambient air directly and continuously.
DIGITEL High Volume Samplers DHA-80 are fully automatic systems to sample dust and aerosol particles for later assessment and analysis (gravimetric and analytical determination). The sampler operation range in standard execution is 100 to 1’500 litres per minute (6 to 90 m³/h).
This instrument provides continuous and reliable measurement of Ozone in ambient air. Measurement is based on the UV photometric method as specified in ISO-13964
DIGITEL Low Volume Samplers DPA-14 are fully automatic systems to sample dust and aerosol particles for later assessment and analysis (gravimetric and analytical determination) in accordance with EN12341:2023
Dasibi 1003-AH Air Water Ozone Concentration PPM Measurement Analyzer Monitor
The EC9852 Sulfur Compounds Analyzer combines microprocessor control with fluorescence detection and an external converter to measure H2S and SO2 or TS TRS and SO2 or H2S SO2 and H2S + SO2 in the range of 0-2 ppm with a detection limit of 0.5 ppb for SO2.
Envea AC32e - NO/NO2/NOx Analyzer (replaces enviroAC32E)
MP101M particulate analyzer (replaces enviroMP101M)
Envea O342e - UV Photometric Ozone Analyzer (replaces enviroO342E)
Environnement S.A. Model 1003-RS under DASIBI license
Environnement SF8 for black smoke and acidity
SINTECH SPECTRAS BTX GC 855 series undetermined
Analysis of VOC/BTEX in ambient air at very low concentration levels (ppb/ppt) thanks to precontration using a trap
SYNTECH SPECTRAS GC 955 series undetermined
Gas chromatograph (GC) specifically built for the measurement of BTEX in ambient air. It can be utilised to simultaneously detect and measure up to 40 different components
SYNSPEC GC955 615/815 Ozone Precursor Analyzer
GRIMM CPC model 5412 Standard Condensation Particle Counter
Compact CPC for nanoparticle counting with optimized detection efficiency and response time with integrated sample flow pump. Different calibrations possible
GRIMM-CPC5420 Scanning Mobility Particle Counter and Sizer
GRIMM model EDM 107 for PM10 PM2.5 PM1 and TC
The GRIMM model 107 monitor is designed to measure particle size distribution and particulate mass based on a light scattering measurement of individual particles in the sampled air
GRIMM model EDM 180-C for PM10 and PM2.5
GRIMM model EDM 180-D for PM10 PM2.5 PM1 and TC
GRIMM model EDM 180-MC for PM10 PM2.5 PM1 TC and 31 size channels
GRIMM model EDM 280 for TSP PM10 PM4 PM2.5 PM1 PMcoarse
GRIMM model EDM 365 for PM10 PM2.5 PM1 and TC
Model 5420 SMPS+C Scanning Mobility Particle Counter and Sizer
IL-Punkkinen for NO2/NO hourly measurements
Device based on the modified automated Saltzman method by Finnish Meteorological Institute.
RA-915AM monitors are designed for continuous mercury measurements in ambient air indoor air and in natural gas. Ideal for permanent unattended installation at air monitoring stations RA-915AM monitor for ambient air allows ultra-sensitive measurements of background Hg concentrations and is compliant with the European standard EN 15852 for ambient air monitoring.The monitors feature fully-automated unattended operation without the need for the monthly technician maintenance and frequent recalibration. The use of the instrument allows cutting down operation costs as the analyzer requires no argon cylinders for operation and no consumables.The remote data transfer makes it possible to connect the monitor to data collection systems via RS232 Modbus and analogue output.
Mercury Analyzer is a portable multifunctional atomic absorption spectrometer with Zeeman background correction which eliminates the effect of interfering impurities. It is the only high sensitivity and selectivity instrument that does not require gold amalgam pre-concentration and subsequent regeneration steps. This enables the user to conduct real time air monitoring and detection of mercury vapor. Being combined with pyrolysis or cold vapour attachments for liquid and solid sample testing the instrument can determine mercury content in water soil food products natural and stack gases may be used for dental and toxicological applications etc.
The MRI Integrating nephelometer is used to measure the scattering coefficient of the air, caused by aerosol particulates.
The PNS DM consists of a low/medium volume sampler (LVS or MVS) and a filter changer with intake tube and sampling inlet. It collects fine particulate matter on sampling filters according to EN 12341:2014 (PM10 and PM2.5).
The PNS T-DM combines a low / medium volume sampler (LVS or MVS) and a filter changer in a stainless steel cabinet. It collects fine particulate matter on sampling filters according to EN 12341:2014 (PM10 and PM2.5).
Passive Palmes Tube. Generally for NO2. However available for other substances like NOX-NO SO2 O3 C6H6...
Picarro G2301 Gas Concentration Analyzer
The Picarro G2301 gas concentration analyzer provides simultaneous, precise measurement of carbon dioxide (CO2) - methane (CH4) at parts-per-billion (ppb) and water (H2O) vapor at parts-per-million (ppm) sensitivity with negligible drift for atmospheric science air quality and emissions quantification. It meets the World Meteorological Organization (WMO) and Integrated Carbon Observation System (ICOS) performance requirements for CO2 and CH4 atmospheric monitoring.
RUPPRECHT & PATASHNICK, SERIES 5400 CARBON MONITOR
The Series 5400 is an automatic speciation analyzer of suspended particulate matter. It measures the elemental and organic carbon contained in suspended particulate matter at averaging times as short as one hour.
The SIR Model S-5001 is a Non-Pulsed U.V. Fluorescence Sulfur Dioxide Analyzer. In the nalyzer optically filtered pure monochromatic U.V. light from a Zinc lamp is focused onto the SO2 reaction chamber.
The SIR Model S-5006 is a Non-Dispersive Infrared (NDIR) Analyzer for the measurement of Carbon Monoxide concentrations in ambient air
The SIR Model S-5012 Nitrogen Oxides Analyzer is designed to measure the amount of nitric oxide (NO) nitrogen dioxide (NO2) and total oxides of nitrogen (NOx) in a continuous ambient air sample
SIR Model S-5014 ozone analyser is a self-contained instrument which measures the concentration of ozone in a sample of ambient background) gas such as air oxygen nitrogen etc
TEOM 1405DF with FDMS Dichotomous monitor for PM2.5 & PM10
The 450iQ analyzer operates on the principle that H2S can be converted to SO2. As the SO2 molecules absorb ultraviolet (UV) light and become excited at one wavelength, the molecules then decay to a lower energy state emitting UV light at a different wavelength.
The Model 3022A Condensation Particle Counter detects particles down to 7 nm in diameter. Using both single-count and photometric detection modes it provides highly accurate measurements for concentrations as high as 107particles/cm3.
The Ultrafine Particle Monitor Model 3031 is specifically designed to measure ultrafine particles long-term. It reports the particle number concentration in six size classes between 20 and 500 nm and requires only minimal maintenance.
TSI's Electrostatic Classifiers 3080 Series produce highly monodisperse, submicrometer aerosol. They have been used in a variety of aerosol-generation or particle-sizing applications with highly accurate and repeatable results.
TSI Nanoparticle Surface Area Monitor 3550
The Nanoparticle Surface Area Monitor 3550 (NSAM) measures the human lung-deposited surface area of particles (reported as µm 2 /cm 3 ) corresponding to tracheobronchial (TB) and alveolar (A) regions of the lung. The NSAM 3550 provides a simple and fast solution for measuring the surface area equivalent dose in the lung. It is an important instrument for research in the fields of inhalation toxicology health effects and epidemiology and for measuring and monitoring workplace exposure.
Enables ambient air monitoring stations to measure size distributions of ultrafine particles in compliance with CEN/TS 17434:2020.
TSI model 3752 High Concentration Condensation Particle Counter
TSI Condensation Particle Counters (CPCs)
Condensation Particle Counter Model 3772 is a compact rugged and full-featured instrument that detects airborne particles down to 10 nm at an aerosol flow rate of 1.0 L/min over a concentration range from 0 to 10000 particles/cm 3. It is ideally suited for applications that do not require measurement of high concentrations.
TSI model 3775 Condensation Particle Counter
TSI model 3775 Condensation Particle Counter (Discontinued)
TSI Ultrafine Condensation Particle Counter (UCPC) Model 3776
The Ultrafine Condensation Particle Counter 3776 (UCPC) a types of condensation particle counter (CPC) is designed primarily for researchers interested in airborne particles smaller than 20 nm. It is a continuous-flow instrument that detects particles down to 2.5 nm. It uses a special sheath-air flow design that improves response to changes in particle concentration and increases counting efficiency for ultrafine particles.
TSI’s EPC™Monitor Model 3783 provides continuous, high precision airborne particle number concentration measurements
General Scanning Mobility Particle Sizer
The Thermo Scientific Model 17i Ammonia Analyzer uses the light producing reaction of nitric oxide (NO) with ozone (O3) as its basic principle. The instrument has three modes of operation NO NOx and Nt.
With ultraviolet photometry technology. High stability and low detection limit (0.5 ppm)
The aeris® AE2010U Trace Level Ozone (03) Analyser employs a non-dispersive ultraviolet (UV) absorption
The aeris® AE2030U Trace Level Carbon monoxide (CO) analyser employs a n on-dispersive infrared (NDIR) photometer to measure very low concentrations of CO
The aeris® AE2041U Trace Level NO/NO2/NOx chemiluminescence analyser incorporates a single-channel design measuring NO and NOx
The aeris® AE2050U Trace Level Sulphur Dioxide (S02) Analyser employs pulsed UV fluorescent radiation technology
The Aerosol Chemical Speciation Monitor (ACSM) measures mass loading and chemical composition of non-refractory aerosol particles in real-time. Smaller lower cost more robust and simpler to operate than our AMS instruments the ACSM is designed for long-term unattended deployment and routine monitoring applications.
Agilent HP 6890 (used for benzene and toluene measurement)
System for analysis of the Ozone precursor VOC
The airmoVOC BTX is an automatic gas chromatograph dedicated to the analysis and monitoring of trace amounts of BTEX compounds(Benzene Toluene Ethylbenzene m- p- and o-Xylenes)
The airpointer‘s modular design comprising a base unit analyzing modules and sensor modules allows for a configurationaccording to different application requirements.
Andersen/GMW Model FH621-N Beta Monitor
Andersen Instruments Model FH62I-N PM10 Beta Attenuation Monitor consisting of the following components: FH101 Vacuum Pump Assembly - FH102 Accessory Kit - FH107 Roof Flange Kit - FH125 Zero and Span PM 10 Mass Foil Calibration Kit - FH62I Beta Attenuation 19-inch Control Module - SA246b PM10Inlet (16.7 liter/min) or the louvered inlet specified in 40 CFR 50 Appendix L Figs. L-2 thru L-19 operated for 24-hour average measurements with an observing time of 60 minutes the calibration factor set to 2400 a glass fiber filter tape an automatic filter advance after each 24-hour sample period and with or without either of the following options: FH0P1 Indoor Cabinet FH0P2 Outdoor Shelter Assembly
Andersen Instruments Model FH62I-N PM10 Beta Attenuation Monitor consisting of the following components: FH101 Vacuum Pump Assembly - FH102 Accessory Kit - FH107 Roof Flange Kit - FH125 Zero and Span PM 10 Mass Foil Calibration Kit - FH62I Beta Attenuation 19-inch Control Module - SA246b PM10Inlet (16.7 liter/min) or the louvered inlet specified in 40 CFR 50 Appendix L Figs. L-2 thru L-19 - operated for 24-hour average measurements with an observing time of 60 minutes the calibration factor set to 2400 a glass fiber filter tape an automatic filter advance after each 24-hour sample period and with or without either of the following options: FH0P1 Indoor Cabinet - FH0P2 Outdoor Shelter Assembly
Thermo Andersen Series FH 62 C14 Continuous PM10 Ambient Particulate Monitor
Thermo Andersen Series FH 62 C14 Continuous PM10 Ambient Particulate Monitor and Thermo Scientific Model 5014i Beta (5014i Beta) Continuous Ambient Particulate Monitor ” operated for 24-hour average measurements with the specified 10-micron EPA PM10inlet (or alternative louvered PM10inlet meeting 40 CFR 50 Appendix L specifications) inlet connector sample tube with heater roof flange kit mass foil kit pump kit sample filter tape with operational settings of 1000 L/h (16.67 L/min) sample flowrate daily filter change auto filter change at volumetric flow <950 L/h auto filter change at mass >1500 micrograms and factory default calculation mode settings operated with software version 1.07. Operated calibrated and serviced according to the appropriate Operator Manual
Andersen/GMW generic High-Volume Air Sampler
3 WAVELENGTH INTEGRATING NEPHELOMETERAerosol particles in the atmosphere directly influence the earth‚Äôs radiative balance by absorbing and scattering the solar radiation and indirectly by changing the clouds microphysical properties."
Bendix/Combustion Engineering Model 8101-C Oxides of Nitrogen Analyze
Bendix or Combustion Engineering Model 8101-C Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range with a Teflon sample filter (Bendix P/N 007163) installed on the sample inlet line.
Columbia Scientific Industries Models 1600
CSI Model 1600 Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range with a Teflon sample filter (CSI P/N M951-8023)installed on the sample inlet line
Columbia Scientific Industries Models 5600
CSI Model 5600 Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range with any signal integration time in the range of 20 to 99 seconds with a Teflon sample filter (CSI P/N M951-8023) installed on the sample inlet line
Environnement S.A. Model AC21M NO2 Analyzer
Environnement S.A. Model AC30M NO2 Analyzer
Environnement S.A. Model AC31M NO2 Analyzer
Environnement S. A. Model AC31M Chemiluminescent Nitrogen Oxide Analyzer operated with a full scale range of 0 -500ppb at any temperature in the range of 15°C to 35°C with a 5-micron PTFE sample particulate filter with the following software settings: Environnement S. A. Model AC31M Chemiluminescent Nitrogen Oxide Analyzer operated with a full scale range of 0 -500ppb at any temperature in the range of 15°C to 35°C with a 5-micron PTFE sample particulate filter with the following software settings: Automatic response time ON Minimum response time set to 60 seconds (RT 2) and with or without any of the following options: Internal Permeation Oven- Connection for Silica Gel Dryer- RS232-422 interface- EV3 valve- Internal Printer
Environnement S.A. Model AC32M NO2 Analyzer
Environnement S. A. Model AC32M Chemiluminescent Nitrogen Oxides Analyzer operated with a full scale range of 0 -500 ppb at any temperature in the range of 10°C to 35°C with a 5-micron PTFE sample particulate filter with response time setting 11 (automatic response time) and with or without the following option:Internal permeation oven.
UV Fluorescence gas monitor offering excellent performances for the monitoring of SO2 in the range 0-10 ppm or 0-1 ppmEco-designed reference air quality monitor QAL1 certified by the TÜV US-EPA approved. Utilizes the standard method for the measurement of sulfur dioxides(EN 14212). With its ultra low power consumption & 24 V supply (option) the AF22e is the sole SO2 monitor on the market able to measure without being connected to the power grid.https://www.envea.global/s/emissions-en/gas-monitors-emissions-en/af22e-dilution/
Total Gaseous Mercury (TGM) Ultratracer. Measuring of Total Gaseous Mercury (TGM) in air and other gases
Environnement S.A. Model AC32E NO2 Analyzer (superseded by EnveaAC32e)
Environnement S.A. Model AF20M SO2 Analyzer
Environnement S.A. Model AF21M SO2 Analyzer
Environnement S.A Model AF21M Sulfur Dioxide Analyzer operated on a range of 0-0.5 ppm with a response time coefficient setting of 01 a Teflon ilter installed in the rear - panel filter assembly and with or without any of the following options: Rack Mount/Slides RS-232-C Interface
Environnement S.A. Model AF22E SO2 Analyzer
Environnement S.A. Model AF22M SO2 Analyzer
Environnement S.A Model AF22M UV Fluorescence Sulfur Dioxide Analyzer operated with a full scale range of 0 - 500 ppb at any temperature in the range of 10°C to 35°C with a 5-micron PTFE sample particulate filter with a response time setting of 11 (Automatic response time) with the automatic “ZERO-REF” cycle ON and set for activation every 24 hours and with or without either of the follow ing options: Permeation oven Rack mount slides
Environnement S.A. Model AS32M Nitrogen Dioxide Analyzer
Environnement S.A. Model CO10M CO Analyzer
Environnement S.A. Model CO12E CO Analyzer
Environnement S.A. Model CO12M CO Analyzer
Environnement S.A Model CO12M Gas Filter Correlation Carbon Monoxide Analyzer operated with a full scale range of 0 - 50 ppm at any temperature in the range of 10°C to 35°C with a 5-micron PTFE sample particulate filter with response time ON and with the automatic “ZERO-REF” cycle either ON or OFF
Environnement S.A. Model CO11M CO Analyzer
Environnement S.A. Model CO11M Ambient Carbon Monoxide Analyzer operated on a full scale range of 0 - 50 ppm at any temperature in the range of 15 °C to 35°C with a 5-micron PTFE sample particulate filter with the following software settings:Automatic nvironnement S.A. Model CO11M Ambient Carbon Monoxide Analyzer operated on a full scale range of 0 - 50 ppm at any temperature in the range of 15 °C to 35°C with a 5-micron PTFE sample particulate filter with the following software settings:Automatic response time ON - Minimum response time set to 40 seconds (RT 13). Automatic ZERO-REF cycle programmed every 24 hours and with or without any of the following options: RS232-422 Serial Interface - Internal Printer
Using a proven flame ionisation detector (FID) the analyzer HC51M measures ambient air hydrocarbons (THC CH4 nmHC) in the range 0-1000 ppm.
Environnement S.A. Model MP101M (may be used for PM10 and PM2.5) (superseded by EnveaMP101M)
Environnement S. A. Model MP101M with PM10 or PM2.5 Beta Gauge Monitor ” configured with the louvered PM 10 or PM2.5 inlet specified in 40 CFR 50 Appendix L or its flat-topped predecessor version and one of the three optional temperature-regulated sampling tubes (RST) and operated with a full scale measurement range of 0 - 0.500 mg/m3 (0 - 500 g/m3) with the sample flow rate set to 1.00 m3/h and flow regulation set to yes the “norms selection” set to m3(actual volume) the “cycle” set to 24 hours the “period” set to none and the “counting time” set to 200 seconds
Environnement S.A. Model MP2.5M undefined
Environnement S.A. Model O341M UV Ozone Analyzer
Environnement S.A. Model O341M UV Photometric Ozone Analyzer operated on a full scale range of 0 - 500 ppb at any temperature in the range of 15°C to 35°C with the response time set to 50 seconds and with or without any of the following options: Internal Ozone Generator - Span External Control - RS232 - 422 Serial Interface - Internal Printer.
Environnement S.A. Model O342E UV Ozone Analyzer (superseded by EnveaO342e)
Environnement S.A. Model O342M UV Ozone Analyze
Environnement S.A Model O342M UV Photometric Ozone Analyzer ” operated with a full scale range of 0 - 500 ppb at any temperature in the range of 10°C to 35°C with a 5-micron PTFE sample particulate filter with response time setting of 11 (Automatic response time) and with or without any of the following options: c) Internal ozone generator d) Span external control (zero/span solenoid valve).
Environnement S.A. SANOA Multigas Longpath Monitoring System
Environnement S.A. Model SANOA Multigas Longpath Air Quality Monitoring System consisting of a receiver one or more projectors interface unit a user-provided control unit computer running the SANOA VisionAIR software and associated incidental equipment configured for measuring SO2 with the temperature control and internal calibration cell options installed operated with a measurement range of 0 to 0.5 ppm over an installed monitoring path length of between 27 and 500 meters within an ambient air temperature range of -30 to +45°C with a measurement (integrating) time of 180 seconds and with or without external temperature and barometric pressure sensors or any of the following options: external (meteo) input connection series 1M bus connection OGR type projector analog outputs.
Horiba model APHA 360E hydrocarbons analyser
Horiba model APHA 360 hydrocarbons analyser
Horiba model APHA 370 hydrocarbons analyser
MCV CAV-A/MSb Captador d'Alt Volum Seqüencial
Meloy Model SA 700 Fluorescence Sulfur Dioxide Analyze
Meloy Model SA 700 Fluorescence Sulfur Dioxide Analyzeroperated on the 0-250 ppb1 the 0-500 ppb or the 0-1000 ppb range with a time constant switch position of either 2 or 3.The analyzer may be operated at temperatures between 20°C and 30°C and at line voltages between 105 and 130 volts with or without any of the following options: FS-1 Current Output - FS-2 Rack Mount Conversion - FS-2A Rack Mount Conversion - FS-2B Rack Mount Conversion - FS-3 Front Panel Mounted Digital Meter - FS-5 Auto/Manual Zero/Span With Status - FS-6Remote/Manual Zero/Span With Status - FS-7 Auto Zero Adjust. voltages between 105 and 130 volts with or without any of the following options: FS-1 Current Output
Multi-gas Micro Monitoring station (MMS) Portable version
Opsis AR500 Open path monitor with ER 110
Opsis AR500 Open path monitor with ER 120
Opsis AR500 Open path monitor with ER 150
Opsis SM200 Particulate Analyser with PM2.5 & PM10 Heads
Osiris air monitoring device measures airborne particles (TSP PM10 PM2.5 and PM1). MCERTS certified for PM10
PHILIPS K50109/00 Gas Filter Correlation CO analyser
PHILIPS K50110/00 UV Photometric O3 analyser
PHILIPS K50206/00 Pulsed Fluorescence SO2 analyser
Picarro G2103 (used for NH3 measurement)
Ecotech serinus 40 used for NOx-NO2-NO modified to measure NOH3 (including a converter)
The Serinus 51 H2S/SO2 instruments can integrate seamlessly into continuous monitoring networks and can sequentially measure H2S and then SO2
EC/OC measurements Sunset Laboratory Inc Thermal / Optical Carbon Analyser (near-real time)
EC/OC measurements Sunset Laboratory Inc Thermal / Optical Carbon Analyser - operated near real time
Tekran mercury analyser 2537A. Elemental mercury is sampled by placing the 1130 before the main 2537A instrument. The respective constituents are sampled from the air using either the 1130 add-on for reactive gaseous mercury or the 1135 add-on for particulate bound mercury.
Thermo model 14B chemiluminescence NO-NO2-Nox
Thermo model 14B/E chemiluminescence NO-NO2-Nox
Gas Filter Correlation CO2 Analyzer 115V-AC
Thermo model 42i NO/NOx analyser - blue light
Thermo model 43h pulsed fluorescence SO2 analyser
Thermo model 43i-TLE (Enhanced Trace Level)
Thermo model 48 CO analyser (gas filter correlation)
Thermo model 48i-TLE (Enhanced Trace Level)
This is a reference photomer not an analyser
The Model 5012 MAAP measures the aerosol-related absorption of light and the corresponding atmospheric black carbon mass concentration. Continuous data is provided by a multiple detector setup to simultaneously measure the transmitted and the scattered light from a particle-loaded filter. Typical applications include determining ambient elemental carbon for health-related effects and aerosol light absorption for visibility monitoring (regional haze) and climate research.
Thermo 5014i Beta Continuous Ambient Particulate Monitor
Measure PM-10 PM-2.5 or PM-1 mass concentrations with the Thermo Scientific™ 5014i Beta Continuous Ambient Particulate Monitor. The 5014i distinguishes itself from other beta measurement methods by utilizing a continuous (non-step wise) mass measurement with a proven industry standard which provides for long-term unattended operation. To accurately address potential water bias and volatile loss the Dynamic Heating System allows the user to hold the sample temperature at a fixed value or below a relative humidity threshold.
Thermo model 5030 SHARP Particular Monitor with PM10 & PM2.5 heads
The Thermo Scientific™ 5030 SHARP continuously measures the mass concentration of particulate utilizing a combination of beta attenuation and light scattering technology.The only continuous (non-step wise) hybrid ambient monitor that provides high resolution measurement of airborne particulates with excellent short and long term precision.