The AGL is an instrument from the UK for measuring BlackSmoke and acidity. The AGL instrument is used in conjunction with the EEL (Diffusion Systems) Model 43 smoke stain reflectometer. The zero is set with the reflectometer switched off. The instrument is checked for correct functioning using a calibrating tile. The measuring head Is set on the white standard tile and the control adjusted to read 100. The head is then moved to the grey half of the tile which has a calibration marked on it. The instrument should agree with this calibration to ±1.5 divisions. The white tile is now covered by a clean Whatman No. 1 filter paper smooth side uppermost. The instrument Is re-set to 100. Stained papers are assessed on the white tile. The instrument does not use a coloured filter.
Instrument for pumping 2m3 air per day through absorbing solution ( in glass container).
Bergerhoff (bulk sampler for deposition of heavy metals and PAHs)
The Digitel High Volume Sampler DH-77 is a system to sample dust and aerosol particles for later assessment and analysis. The sampler operation range in standard execution is 100 to 1000 litres per minute (6 to 60 cubic metres per hour). The system's usual designation is 'High Volume Sampler'. Air-borne-dust parts in the sampled air are separated onto 150 mm diameter filters
INTERSCIENCE Finnigan trace GC ultre + trace DSQ
The IHE instrument is an instrument from Belgium to measure BlackSmoke and acidity. The IHE instrument is used with the Diffusion Systems Reflectometer Mark III Model 11. The instrument is zeroed using a standard black tile and the span (100) by setting the measuring head on the smooth side of a Whatman No. 1 filter paper set on a standard white tile - supplied with the black standard. The smoke stained filter is evaluated on the standard white tile. The reflectometer uses a green filter.
Medium volumen sampler for PM10 and PM2.5
Low volume sampler for gravimetric determination of the mass concentration PM10 or PM2.5 of particulate matter in suspension, with sample collection on filter and determination in laboratory. With a capacity of up to 30 filters
System for the collection of samples in absorption tubes for subsequent analysis in the laboratory. The instrument is intended for the collection of volatile organic compounds present in the atmosphere. The collection is carried out in tubes with chemical sorbents, with daily sampling, which allow its collection. The equipment has a sequential module to be able to carry out continuous campaigns of several days
High Volume Sampler HVS16. Microcomputer controlled gas collector for the automated dust sampling on membrane filters with a diameter of 140 mm or 150 mm in acc. with EN 12341. Automatic switching to sequentially admit up to 16 filters
Collection on 16 filters for the subsequent laboratory analysis in samples.
Control system for the automated collection of gases and particles on adsorption tubes, wash-bottles, or filters. The control program has a modular structure and, with an optional sampling switching, can log up to 8 sequential samples (optionally up to 16)
EIGENBRODT (NSA 181 KHT) - WET ONLY SAMPLER
ENVIRONNEMENT AUTOMATIC SEQUENTIAL PARTICULATE SAMPLER
PM162M is a sequential automatic sampling device for particulate matter in the ambient air. The particles are sampled at constant flow rate (1 m3/h or 2.3 m3/h) and collected on filters for gravimetric weighing and possible physicochemical laboratory analysis
Sequential Air Sampler, Type EK, Aerosol filter as for sulphate + oxalic acid im
Sequential Air Sampler, Type EK. KOH-impregnated Whatman 40 filter
Sequential Air Sampler, Type SS2000. NaI-impregnated glass sinters
Diffusion Systems Ltd: Smoke Stain Reflectometer mod. 43
This will be remove shortly from CODELIST
Digitel DPA96MV (active sampling for VOC)
The Environnement SF8 is an instrument from France to measure Black Smoke & Acidity The Environnement SF8 instrument is used with a Photovolt Model 670 reflectometer, to measure the degree of staining on Schneider CA 32 filter paper. The reflectometer is zeroed by setting the measuring head on a standard black tile and to 100 when set on a pile of clean Schneider filter papers. A standard white tile of similar reflectance to the filter papers is used to check the calibration which should not differ from clean filter paper by more than ±2 divisions. Each stain is measured with the stained paper set on a pile of clean filter papers. The reflectometer uses a green optical filter.
Modified diffusion tube by Gradko to measure NH3
The LIFETEK PMS model is certified for outdoor use in temperatures ranging from -20 C° to 50 °C. https://dilus.com/wp-content/uploads/2021/02/LIFETEK-PMS-SERIES-1.pdf
The Ogawa sampler is a badge-type sampler that can be equipped with a sampling filter on either side of the sampler body. Sampling filters consist of cellulose fibre.
To include both PM10, PM2.5 or other PM fraction...
Passive sampler (Carbopack E),Perkin-Elmer ATD 400,HP GC 5890,HP MS 5792
PERKIN ELMER Turbomatrix Desorber, PERKIN ELMER Clarus 500
Semi-automatic combined filter and impinger
Semi-automatic (combined filter and impinger) device for daily sampling of NO2, SO2, and soot in ambient air
Sunset Laboratory Inc Thermal / Optical Carbon Analyser
Thermo Scientific: Trace GC ultra + DSQII
Thermo Scientific: Trace GC ultra + Trace MS
The PNS DM consists of a low/medium volume sampler (LVS or MVS) and a filter changer with intake tube and sampling inlet. It collects fine particulate matter on sampling filters according to EN 12341:2014 (PM10 and PM2.5).
The PNS T-DM combines a low / medium volume sampler (LVS or MVS) and a filter changer in a stainless steel cabinet. It collects fine particulate matter on sampling filters according to EN 12341:2014 (PM10 and PM2.5).